BAYER Contour Control Solution High Level
The Bayer Contour Control Solution is a 2.5 mL desired aqueous glucose solution for verifying the accuracy of the Bayer Contour blood glucose test Strips. Bayer Contour Control Solution ensures that your Bayer Contour meters and test strips are working properly. The product contains a known amount of glucose that reacts with the glucose test strips.
- BAYER Contour Control Solution High Level – 2.5mL glucose control solution
- Instructions for using Bayer Contour control solution
The Bayer Contour Control Solution is designed to check the accuracy of Bayer contour glucose meters and test strips. Simply apply a drop of the Contour glucose control solution to a test strip and run a standard glucose test. Confirm that the test results are within the accepted range printed on the test strip vial. It is advised to test your glucose meter and strips accuracy on a regular basis.
When to Perform a Control Solution Test
Bayer Contour Control solution can be used to verify that your Bayer Contour Glucose Monitoring System is working correctly and giving accurate results. The system is performing correctly if the control solution test result falls within the specific control solution range listed on your test strip vial. You are encouraged to use Control Solution on a regular basis, specially with each new vial of blood glucose test strips, after changing meter batteries, or if you suspect your meter is delivering false results.You should also use control solution before using your glucose monitoring system for the first time or if the test strip vial has been left open or exposed to extreme heat, cold or humanity. For further details please contact the manufacturer support.
Store your Bayer Contour Control solution at 59-86 degrees F, Do Not Refrigerate, Discard after 3 Month After Opening, For In Vitro Diagnostic Use Only.